Medical Device Field Correction - April 2015
Product: Invacare® Pronto® Air Personal Transporter
Model numbers: AIRPTMBDY18, AIRPTMBDY20, AIRPTBASE (Service Part Number: LPC12-18)
Date Recall Launched: April 2015
Invacare has recently identified a quality issue with the batteries for use in the Invacare® Pronto® Air Personal Transporter. There is a possibility that the batteries could overheat during charging. The batteries may swell or leak as a result of overheating. Swollen batteries may become non-functional and contact with leaked battery acid may result in chemical burns or property damage. Not all units identified as impacted will show signs of battery deformation.
Medical Device Field Correction - January 2015
Product: Invacare® FDX®, TDX® series and Storm Series® power wheelchairs with 20"-24" width configurations with Formula CG Recline seating system (i.e. Tilt/Recline, Tilt/Recline/Elevate, or Recline only) and a conventional style back
Date Recall Launched: January 16, 2015
Invacare has recently identified a potential pinch point affecting the wiring involving Invacare® FDX®, TDX® series and Storm Series (3G)® power wheelchairs with 20"-24" width configurations with Formula CG Recline seating system (i.e. Tilt/Recline, Tilt/Recline/Elevate, or Recline only) and a conventional style back. There is a possibility that if the slack in the wires is not routed and secured correctly, and with flexing of the back pan under the user's weight, the headrest knob or clamp bracket may pinch, damage or cut the wire creating a hazard. Smoking, sparking, burning or fire may occur if any wire in this area is repetitively pinched at the same location over a period of time. Not all impacted chairs are expected to have slack in the wiring that is not routed and secured correctly.
Medical Device Field Correction - Dianz Capacitor - December 2014Product: Component- Dianz® Capacitor (Manufactured by Xindian Part number 1099804) used in the Invacare® PerfectO2(TM) Oxygen Concentrator. No service parts were impacted in the U.S.
Model numbers: IRC5P & IRC5PO2
Invacare has recently identified a supplier quality issue that potentially impacts certain Invacare® PerfectO2(TM) concentrators. Invacare has learned that the PSI (Pressure Sensitive Interrupter) design for the compressor motor start/run capacitor by one supplier could lead to rupture.
For service or more information regarding field corrections, please contact your local Invacare dealer. To locate a dealer in your area, utilize our Dealer Locator or call Invacare: US - 800-333-6900 or Canada - 800-668-5324 to speak with a customer service representative.